You can do more than just check that it appears in MAX. When I use the “search for instruments” function under tools and instrument drivers , LabView does not detect the Keithley. Before running the VI type in the desired file name. The combination of power and precision allows you to use the same instrument for both high power sweeps and low current measurements; while the addition of a high-speed update rate and sampling rate allow you to use the instrument in non-traditional ways, such as generating and measuring a waveform. When writing an instrument control application for a serial instrument, you will communicate with a COM port rather than the serial instrument connected to the COM port. I am currently studying how to control the voltage and the current of a source meter which is Keithley using Labview software. Place the unplaced controls and indicators down onto the Panel.
|Date Added:||20 January 2005|
|File Size:||7.44 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Use the following procedure to add manual inputs for Voltage, Current, and Resistance ranges Create an “Enable Auto Range” boolean control by right clicking the “Enable Auto Range” boolean constant and selecting Change to control. We want users to be able to choose to have the Auto Zero enabled, disabled, or run immediately so we will create a Kiethley at the Autozero terminal.
Open the Panel by selecting Panel in the View Selector. Open the Learning tab by selecting Learn to Program tile labfiew the Welcome screen or select the Learning tab from the top right.
Sometimes instrument drivers default to certain settings you may not desire and may not even know about until you start digging. I have keitgley idea what to do or how to fix this.
On the Write Delimited Spreadsheet Node create a constant for the top keitlhey terminal. I am trying to do some very basic measurements with a Keithley Sourcemeter on LabView.
Reposition controls on the Diagram for Configure Output and Enable Output nodes to make room for extra controls. You can do more than just check that it appears in MAX. When writing an instrument control application for a serial instrument, you will communicate with a COM port rather than the serial instrument connected to the COM port.
usb – How to use labview to control the voltage and current of Keithley – Stack Overflow
The instrument is now configured. How do I properly verify that there is a correct connection?
Joe Friedrichsen 1, 12 Find the section of the Diagram between the third Configure Measurement node and the Configure Output node. Become a Redditor and subscribe to one of thousands of communities.
Seki 9, 5 33 First ensure you have the NI-Serial driver installed.
You can then place items down one by one on the Panel. Rename each control to match the measurement that’s being configured, “Voltage Manual Range” for example. Also, I’ve already checked the serial configuration and they all match.
Keithley 648X Series Meter
Create a control for the name or relative path terminal on the left hand side of the Build Path Node. The instrument will then be detected.
Keithley X Series Meter – IEEE (GPIB), Serial Driver for LabVIEW – National Instruments
This combination of features allows you to use NI system SMUs in a wide range of applications; from materials research and parametric test to high volume production test of RF and mixed signal ICs. Submit a new link. Back to Top 2.
I am currently studying how to control the voltage and the current of a source meter which is Keithley using Labview software. Do the same for actual maximum and actual minimum terminals on the right hand side of the Histogram Node. Submit a new text post. Labvlew Undefined Header”. How do I get setup properly?